Infratec 1241 is a whole grain analyzer. These devices use near-infrared spectroscopy to test multiple parameters in a wide range of oilseeds, legumes and cereals. This is a fast and extremely-precise method.
A huge number of grain processing companies use this device around the world to assess grain quality. Unique calibrations based on extensive data make it rapid, accurate and versatile analysis device. In addition, there are additional modules that allow to analyze the flour and test weight.
|Grains, oilseeds, beans and pulses – they can all be tested quickly, accurately and easily. Most grains and oilseeds can be tested without any preparation.||Moisture, protein, oil, test weight and many other parameters.|
ADVANTAGES OF THE ANALYZER INFRATEC 1241
Infratec 1241 has the highest accuracy and is easy to use, which allows it to be used in various businesses. The main principle behind the Infratec is that it should be easy to use for any type of business. This task is achieved thanks to ready-to-use ANN calibrations as well as the high quality of the device and qualified staff. The easy transferability of information between the units make it possible to establish a network of different analyzers for quality and speed assurance.
THEREFORE, GRAIN ANALYZER INFRATEC 1241 IS:
- Quick and simple whole grain analysis without sample preparation.
- Reliable analysis under a variety of conditions.
- Transferability of the calibrations and the ability to establish a network.
These devices analyze such grains as:
wheat, durum wheat; maize; barley; lupin; malt; peas; oats; beans; rye; lentils; triticale; etc.
In addition, the Infratec 1241 analyzes milling products such as:
wheat flour; corn meal; semolina; rice meal; soy meal and ground wheat.
This device analyzes oilseedssuch as rapeseed, sunflower, soybean, etc. as well.
Such analyzers will undoubtedly be useful for plant breeders and seed companies. This is due to the fact that it is possible to work with small amounts of sample, analyzing the color and so on. This devise can also be used in production of beer and biofuel. This is because the materials for the manufacturing such as malt, green malt, barley, etc. can easily be analyzed.
APPLICATION IN OTHER BRANCHES
This device is also suitable for the analysis of alcohol, whiskey, beer, rice products and so on.
The Infratec 1241 is part of a modular system, which also includes a flour module, a test weight module and sample transport module to analyze small amounts of dry samples, wet samples and liquids.
STANDARDS FOR MEASUREMENT RESULTS
These analyzers are officially approved and installed all over the world as the main standard for analysis of protein, moisture, oil and starch in barley, wheat and other grains and oilseeds. We supply the technology that is approved by a great variety of businesses as well as the governments.
- large number of samples;
- it is possible to analyze the entire kernel, not just the surface;
- modulation of near infrared signals;
- deeper penetration;
- low influence of particle size.
Regardless of the installation conditions, you can simply plug in the device, then turn it on and after a short self-test you can produce accurate results. It should be noted that the accuracy of the results does not depend on samples and the temperature change. Therefore, you will get correct results under all conditions.
|No. of sample||Results Infratec 1241, at the temperature of||Lab. results|
|-5 C0||+5 C0||+24 C0||+40 C0||+45 C0|
|Dimensions (W x D x H)||500 × 570 × 400 mm|
|Voltage||AC220-240V 50-60 Hz or AC110-120V 50-60 Hz|
|Rated current||1.0A (110-120V) / 0.5A (220-240V)|
|Wavelength range||570-1100 nm|
|Optical bandwidth||7 nm|
|Number of data points||265|
|Light source||Halogen lamp|
|Storage||Flash disk, USB memory stick|
|Display||640 × 480 TFT LCD|
SAMPLING AND RESULT PRESENTATION
|Analysis time||50 seconds for 10 sub-samples|
|Path length||Variable cell automatically adjusted from 6-33 mm|
|Result report||Presented on the display as default. Can be sent to PC/LIMS and the printer port.|
|Outlier function||Warnings and options for the presentation of the result|
|Regression programs||ANN (Artificial Neural Network); PLS (Partial Least Squares)|
|Number of sub-samples||1-20|